the column cylinder equipped with outside heating mantle, rotor and inside condenser. The fixing frame of rotor is precisely equipped with scratch film and anti-splash device. The inside condenser is placed in the center of evaporator, and the rotor should rotor between column cylinder and condenser.
The material should add from the top of evaporator and continuously spread to the heating surface through the material covered on rotor. Then, the scratch film is capable of scratching the material into a thin and flowing liquid film. During this process, the light molecular exceeded the heating surface form the condense liquid on inside condenser without any collision. Then, the molecular flows along the condenser tube, and discharge through the discharge tube on the bottom of evaporator. The residue liquid (heavy molecule)is collected on the circular channel under the heating zone, and flows through the discharge tube on one side.
Short Path Distillation Parts
Glass parts are connected by frosted mouth,easy to operate and good sealing
Adopt two ports reaction bottle,with a special feeding port,convenient for feeding
Materials can be extracted continuously during the experiment without pressure relief
Micro vertical spine distiller, vertical spine structure, used for fractionation function;Vacuum sandwich is used to weaken the influence of wind and greenhouse to maintain the appropriate temperature gradient in the column
|Heating temperature（℃）||Room temperature -250|
Diaphragm vacuum pump